Digital input DI and digital output DO: The wizard has finished the installation of the driver software, and the USB is ready for use, after you click Finish Norwegian: For additional information, see the Global Shipping Program terms and conditions – opens in a new window or tab. The DAQmx palette on the Functions palette. Differential coupling is of course more robust against differences in grounding potentials, so it should be the default coupling type. Depending of the range of the resistor to measure , you just need to connect it to GND.
|Date Added:||19 September 2014|
|File Size:||50.85 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
The DAQmxBase Read function reads the voltage value of the analog input channel as defined in aitask. This is because the function can be used to read several inputs simultaneously.
You can use the Index Array function on the Array palette. If you know what an ni daq usb 6008 impedance is, you should be able to do the complex math to match your 608. To see the details about a virtual channel, you can click the Show Details buttoncf.
When the wizard has finished the dsq of the driver software, the USB is ready for use after you click Finish Norwegian: I have done that in the past but forgot when I posted earlier in ni daq usb 6008 thread.
Here, let us test analog output channel 0 AO0 and the analog input channel 0 AI0 to see if they work correctly. Only some math involved.
I think that to get more accurate results the shunt resistor should be similar to our load value right? If for some reason AI0 detects some other voltage than the value we set for AO0, then there is an error in either the AI0 channel or in the the AO0 channel, and ni daq usb 6008 investigations are necessary. In general, a task can contain ni daq usb 6008 or more global or local virtual channelsand one such virtual channel is based on one physical channel having e.
See all condition definitions – opens in a new window or ji A local virtual channel exists only within it’s parent task, while a global virtual channel may be used in several tasks. Back to home page. Message 3 of Then, you should test the individual channels of the USB to check that the input signals are detected correctly by the USB, and that the output signals generated by the USB have correct values.
ni daq usb 6008
Greetings from Germany Henrik LV since v3. We may then repeat this procedure for other channels. The Voltage Output dialog window in the Test Panels window. The New Data Acquisition Device dialog uwb.
Use two analog input channels to measure the voltages across both resistors. As you can see form those values, I will have to use 3 or 4 wire connection in measuring impedance due to the high values of electrode resistance and contact impedance in textiles. On-board voltage sources available at individual terminals: See Details – opens in a new ni daq usb 6008 or tab. The following relevant documents can be found at http: In this example, no feedback control system is implemented.
measurement of resistance by USB – Discussion Forums – National Instruments
In general, differential signal coupling is preferrable 608 single ended coupling due to better DC noise suppression. Finn Haugen finn techteach. Special financing available Select PayPal Credit at checkout to have the option to pay over time. Above we created a task for an ni daq usb 6008 input.
Interactive Control Panel for USB-6501, USB-6008, and USB-6009 using DAQmx Base
Digital input DI and digital output DO: Figure 7 shows MAX. However, the Ni daq usb 6008 has several input channels and several output channels. Depending on the range of the unknown impedance and the test current you want to apply, you maybe need an amplifier for the AO